SDRPlay Independent Community Forum › Forums › SDR Software (RSP Compatible) › Spectrum Analyser setting the reference level › Reply To: Spectrum Analyser setting the reference level
July 3, 2020 at 4:53 am #1182
The file Ref-Level-50dBm.png was attached in error. It may be of interest though. It’s from the manual and shows the affect of reference level settings on the noise floor with an input level of -50dBm. Check the manual for information regarding each trace.