Rick, thanks much. The instruction about clicking on the S-meter to change it to a center-tuning meter was buried in a paragraph about signal strength standards in the manual, and I had missed it on the first pass.
I’ve seen the error displayed in synchronous AM, but I need it for NFM. The center-tuning meter shows it clearly, now that I know how to invoke it, but I’d need a way to actually quantify it with high resolution, and export it as data somehow. Alas, this may have to wait for a plug-in from someone with skills I don’t yet have.
73 – Paul AE4KR
- This reply was modified 2 years, 4 months ago by .